|4/2016 - 9|
View TOC | « Previous Article | Next Article »
A Novel Non-Iterative Method for Real-Time Parameter Estimation of the Fricke-Morse ModelSIMIC, M. , BABIC, Z. , RISOJEVIC, V. , STOJANOVIC G. M.,
|View the paper record and citations in|
|Click to see author's profile in SCOPUS, IEEE Xplore, Web of Science|
|Download PDF (1,272 KB) | Citation | Downloads: 872 | Views: 2,678|
bioimpedance, biological system modeling, estimation, filters, signal processing
measurement(10), bioimpedance(9), physiological(7), impedance(5), time(4), measurements(4)
Blue keywords are present in both the references section and the paper title.
About this article
Date of Publication: 2016-11-30
Volume 16, Issue 4, Year 2016, On page(s): 57 - 62
ISSN: 1582-7445, e-ISSN: 1844-7600
Digital Object Identifier: 10.4316/AECE.2016.04009
Web of Science Accession Number: 000390675900009
SCOPUS ID: 85007622286
Parameter estimation of Fricke-Morse model of biological tissue is widely used in bioimpedance data processing and analysis. Complex nonlinear least squares (CNLS) data fitting is often used for parameter estimation of the model, but limitations such as high processing time, converging into local minimums, need for good initial guess of model parameters and non-convergence have been reported. Thus, there is strong motivation to develop methods which can solve these flaws. In this paper a novel real-time method for parameter estimation of Fricke-Morse model of biological cells is presented. The proposed method uses the value of characteristic frequency estimated from the measured imaginary part of bioimpedance, whereupon the Fricke-Morse model parameters are calculated using the provided analytical expressions. The proposed method is compared with CNLS in frequency ranges of 1 kHz to 10 MHz (beta-dispersion) and 10 kHz to 100 kHz, which is more suitable for low-cost microcontroller-based bioimpedance measurement systems. The obtained results are promising, and in both frequency ranges, CNLS and the proposed method have accuracies suitable for most electrical bioimpedance (EBI) applications. However, the proposed algorithm has significantly lower computation complexity, so it was 20-80 times faster than CNLS.
|References|||||Cited By «-- Click to see who has cited this paper|
| H. Fricke, S. Morse, "The electric resistance and capacity of blood for frequencies between 800 and 4(1/2) million cycles," The Journal of General Physiology, vol. 9, no. 2, pp. 153-167, 1925. |
[CrossRef] [SCOPUS Times Cited 154]
 K. S. Cole, "Permeability and impermeability of cell membranes for ions," Cold Spring Harbor Symposia on Quantitative Biology, vol. 8, pp. 110-122, 1940.
 D. Meroni et al., "Healthy and tumoral tissue resistivity in wild-type and sparc-/- animal models," Medical and Biological Engineering and Computing, pp. 1-9, 2016.
[CrossRef] [Web of Science Times Cited 6] [SCOPUS Times Cited 7]
 B. Sanchez, A. S. Bandarenka, G. Vandersteen, J. Schoukens, R. Bragos, "Novel approach of processing electrical bioimpedance data using differential impedance analysis," Medical Enginerring and Physics, vol. 35, no. 9, pp. 1349-1357, 2013.
[CrossRef] [Web of Science Times Cited 27] [SCOPUS Times Cited 29]
 B. Sanchez, E. Louarroudi, E. Jorge, J. Cinca, R. Bragos, R. Pintelon, "A new measuring and identification approach for time-varying bioimpedance using multisine electrical impedance spectroscopy," Physiological Measurement, vol. 34, no. 3, pp. 339-357, 2013.
[CrossRef] [Web of Science Times Cited 57] [SCOPUS Times Cited 59]
 B. Sanchez, E. Louarroudi, R. Pintelon, "Time-invariant measurement of time-varying bioimpedance using vector impedance analysis," Physiological Measurement, vol. 36, no. 3, pp. 595-620, 2015.
[CrossRef] [Web of Science Times Cited 17] [SCOPUS Times Cited 18]
 S. Prakash et al., "Ex vivo electrical impedance measurements on excised hepatic tissue from human patients with metastatic colorectal cancer," Physiological Measurement, vol. 36, no. 2, pp. 315-328, 2015.
[CrossRef] [Web of Science Times Cited 34] [SCOPUS Times Cited 37]
 A. G. Gheorghe, C. V. Marin, F. Constantinescu, M. Nitescu, "Parameter Identification for a New Circuit Model Aimed to Predict Body Water Volume," Advances in Electrical and Computer Engineering, vol. 12, no. 4, pp. 83-86, 2012.
[CrossRef] [Full Text] [Web of Science Times Cited 4] [SCOPUS Times Cited 5]
 P. Van Liedekerke, M. Palm, N. Jagiella, D. Drasdo, "Simulating tissue mechanics with agent-based models: concepts, perspectives and some novel results," Computational Particle Mechanics, vol. 2. no. 4, pp. 401-444, 2015.
[CrossRef] [Web of Science Times Cited 134] [SCOPUS Times Cited 150]
 M. T. Wilson, M. Elbohouty, L. J. Voss, D. A. Steyn-Ross, "Electrical impedance of mouse brain cortex in vitro from 4.7 kHz to 2.0 MHz," Physiological Measurement, vol. 35, no. 2, pp. 267-281, 2014.
[CrossRef] [Web of Science Times Cited 6] [SCOPUS Times Cited 9]
 T. J. Freeborn, B. Maundy, A. S. Elwakil, "Extracting the parameters of the double-dispersion Cole bioimpedance model from magnitude response measurements," Medical and Biological Engineering and Computing, vol. 52, no. 9, pp. 749-58, 2014.
[CrossRef] [Web of Science Times Cited 54] [SCOPUS Times Cited 63]
 Y. Yang, W. Ni, Q. Sun, H. Wen, Z. Teng, "Improved Cole parameter extraction based on the least absolute deviation method," Physiological Measurement, vol. 34, no. 10, pp. 1239-1252, 2013.
[CrossRef] [Web of Science Times Cited 26] [SCOPUS Times Cited 32]
 I. Nejadgholi, H. Caytak, M. Bolic, I. Batkin, S. Shirmohmmadi, "Preprocessing and Parameterzing Bioimpedance Spectroscopy Measurements by Singular Value Decomposition," Physiological Measurement, vol. 36, no. 5, pp. 983-999, 2015.
[CrossRef] [Web of Science Times Cited 7] [SCOPUS Times Cited 10]
 M. Simic, Z. Babic, V. Risojevic, G. Stojanovic, A. A. Ramos, "System for Rapid and Automated Bioimpedance Measurement," In Proceedings of the SDPS Conference, Dallas, Texas, USA, pp. 242-247, 2015.
 S. Rossi et al., "A low power bioimpedance module for wearable systems," Sensors and Actuators A: Physical, Vol. 232, pp. 359-367, 2015.
[CrossRef] [Web of Science Times Cited 12] [SCOPUS Times Cited 15]
 S. Kaufmann, A. Malhotra, G. Ardelt, M. Ryschka, "A high accuracy broadband measurement system for time resolved complex bioimpedance measurements," Physiological Measurement, vol. 35, no. 6, pp. 1163-1180, 2014.
[CrossRef] [Web of Science Times Cited 12] [SCOPUS Times Cited 14]
 P. J. Soh, G. A. Vandenbosch, M. Mercuri, D. M. P. Schreurs, "Wearable wireless health monitoring: Current developments, challenges, and future trends," IEEE Microwave Magazine, vol. 16, no. 4, pp. 55-70, 2015.
[CrossRef] [Web of Science Times Cited 155] [SCOPUS Times Cited 187]
 J. Wang, J. Liang, F. Gao, L. Zhang, Z. Wang, "A method to improve the dynamic performance of moving average filter-based PLL," IEEE Transactions on Power Electronics, vol. 30, no. 10, pp. 5978-5990, 2015.
[CrossRef] [Web of Science Times Cited 101] [SCOPUS Times Cited 125]
 J. Ferreira, F. Seoane, A. Ansede, R. Bragos, "AD5933-based Spectrometer for Electrical Bioimpedance Applications," Journal of Physics: Conference Series, vol. 224, no. 1, pp. 012011, 2010.
[CrossRef] [SCOPUS Times Cited 34]
 E. W. Karas, S. A. Santos, B. F. Svaiter, "Algebraic rules for computing the regularization parameter of the Levenberg-Marquardt method," Computational Optimization and Applications, pp. 1-29, 2016.
[CrossRef] [Web of Science Times Cited 19] [SCOPUS Times Cited 13]
Web of Science® Citations for all references: 671 TCR
SCOPUS® Citations for all references: 961 TCR
Web of Science® Average Citations per reference: 32 ACR
SCOPUS® Average Citations per reference: 46 ACR
TCR = Total Citations for References / ACR = Average Citations per Reference
We introduced in 2010 - for the first time in scientific publishing, the term "References Weight", as a quantitative indication of the quality ... Read more
Citations for references updated on 2023-06-09 10:56 in 109 seconds.
Note1: Web of Science® is a registered trademark of Clarivate Analytics.
Note2: SCOPUS® is a registered trademark of Elsevier B.V.
Disclaimer: All queries to the respective databases were made by using the DOI record of every reference (where available). Due to technical problems beyond our control, the information is not always accurate. Please use the CrossRef link to visit the respective publisher site.
Faculty of Electrical Engineering and Computer Science
Stefan cel Mare University of Suceava, Romania
All rights reserved: Advances in Electrical and Computer Engineering is a registered trademark of the Stefan cel Mare University of Suceava. No part of this publication may be reproduced, stored in a retrieval system, photocopied, recorded or archived, without the written permission from the Editor. When authors submit their papers for publication, they agree that the copyright for their article be transferred to the Faculty of Electrical Engineering and Computer Science, Stefan cel Mare University of Suceava, Romania, if and only if the articles are accepted for publication. The copyright covers the exclusive rights to reproduce and distribute the article, including reprints and translations.
Permission for other use: The copyright owner's consent does not extend to copying for general distribution, for promotion, for creating new works, or for resale. Specific written permission must be obtained from the Editor for such copying. Direct linking to files hosted on this website is strictly prohibited.
Disclaimer: Whilst every effort is made by the publishers and editorial board to see that no inaccurate or misleading data, opinions or statements appear in this journal, they wish to make it clear that all information and opinions formulated in the articles, as well as linguistic accuracy, are the sole responsibility of the author.